NEOTHERM™ -NEONATAL FULL BODY COOLING SYSTEM
Rectal Probe Use
Restart from Power Failure
- Dimension: L 460mm, H 530mm,W 530mm
- Weight: 17kg (Note: weight when reservior is full of Fluid)
- Electricity Consumption:-1.5A/230 VAC,3.1 A/110 VAC, Max. 345 W
- Noise output :-54-54.5dB (Mean value 1m distance)
- Fluid capacity: 350ml (Inner Tank). 500ml (Fill Up Set)
- Fluid active ingredients: Water
- Minimum Fluid Temperature :-12°C
- Maximum Fluid Temperature :-39°C
- Maximum Patient Weight :- 10kg
- Neotherm Neo Starter kit:- TS-NEO-SK
- Includes:- Neotherm™ Neo Mains Cable
- 1 Rectal Probes(Reusable)
- 1Skin Probes(Reusable)
- 2m Mattress Connecting Hoses
- Fill-Up Set
- Inter Layer Foils (Case of 10)
- 1 Aqua Wrap Around Mattress complete with ties (Reusable)
How NEOTHERM™ -NEONATAL FULL BODY COOLING SYSTEM Works ?
HYPOTHERMIA THERAPY :-
There is one established treatment that can minimize permanent brain damage from hypoxic-ischemic encephalopathy (HIE). However, it must be given very shortly after birth/the oxygen depriving incident in order to be effective (Ideally within six hours). This treatment is known as hypothermia therapy, but it has many other names, such as “therapeutic hypothermia,” “cooling therapy” and “neonatal cooling.“ Hypothermia therapy involves cooling the baby down to a temperature below homeostasis to allow the brain to recover from a hypoxic-ischemic Injury. Typically, the target temperature is about 33.5 degrees Celsius (92.3° Fahrenheit)
NEO COOLING :-
Is a closed-loop control system consists of a mattress that is wrapped around the baby’s trunk and legs.The device is fully automated and requires no further input after initiation. The baby temp is cooled to 33.5°C within 25 minutes. This temperature is maintained for 72 hours and a fully automated re-warming phase starts.The re-warming phase occurs to 37°C at 0.02 to 0.5°C/hour.
INSTRUCTIONS TO USE :-
A rectal probe is inserted into the patient (approximately 3 to 5cm) and the temperature is feded back to the Neotherm™.Using thermoelectric cooling module the temperature of the cooling fluid is controlled by varying the amount and direction of electric current.This increases and decreases the fluid temperature by small amounts.The fluid is then passed round the infant using a mattress or a wrap returned to the Neotherm™ The thermal properties of the fluid make it extremely efficient at heat transfer and the Neotherm™ uses surprisingly little energy to maintain a constant temperature.
Simple alarms in the Neotherm™ ensure that users and nursing staff are drawn to the device only when there is a problem. These alarms include:
- Low fluid
- Low Flow
- Temperature Out of Range
You can transfer One Therapy Data upto 80 hours to system using softare.
MODES OF OPERATION:-
Servo-Controlled Rectal temperature
The Neotherm™ Neo maintain the temperature that Clinician has set via the rectal temperature probe.Rate of change can also be set allow even re-warming and cooling.
The Neotherm™ has a complete range of accessories.Mattresses and Wraps with lining sheets for contact with the baby,rectal and skin probes,surface water Connecting Hoses,Fill-Up Set.
Data Storage for allws simple analysis of temperature profiles | Large colour screen is easy to see across the room | benchtop mountable for space saving small/Lightweight potable device | restarts from same settings in event of power failure | USB output | Optional use of skin pobes.
HOW IT WORKS :-
-It should induce cooling rapidly to desired core temperature.
-It should maintain the core temperature within the target range.
-It should allow re-warming in a slow and controlled manner.
-It should be easy to use.
-It should required minimal nursing input.
-It should not interfere with access to the baby.
-Smooth, linear re-warming.
-Compatible with reusable accessories.